Centralization of tools
All the auxiliary instruments to the microtomy process are grouped in a single management, with the centralization of all the information of the process in REAL TIME.
Centralization of alarms
Anomaly and warning alarms, the software isolates and indicates the origin of the anomalies.
Remote diagnosis of the machine and intervention on the software to correct in a timely manner any anomalies and problems
Marconi manages a bidirectional exchange of the orders; it is possible to receive, from a MES system, the production orders. The machine calculates the order and returns it to the MES, with the real production executed.
The cold plate in Teflon, maintains a constant temperature between -16/18°, The histological specimen is placed on this plate in prismatic form, and then inserted into the microtome (cutting system not managed by the bench) which performs the sectioning.
The drying bath maintains a constant temperature of 42/43°C on which the sections of interest of the histological specimen are placed. These are taken with a histological slide and placed in the hot plate (39/40°C) to evaporate excess water.
Man/machine interface where are centralized in
REAL-TIME all the information concerning the process;
it is possible to parameterize all the process utilities, visualize the status of the utilities, the alarms and the system warnings
Sucks up scraps consisting of solid kerosene crumbs.
The suction filter is housed away from heat sources to avoid melting of the kerosene preventing damage to the motor system.
Latest generation process controller allowing both automation and high level data management through the latest protocols available on the market.
Adjustable monitor arm
Control Panel (HMI)
Connections area and buttons
Safety reset button
Power on switch
Waste collection box
Kerosene flake box
Bench power supply